The pocket-sized RaySafe DXR+ operates down to 30 kVp and gives an objective, reproducible and immediate read-out.
- Fully automatic
- Radiographic and Mammography
- Ideal for digital imaging
- 6 - 8 years battery life
“The DXR is amazing – Before I had multiple steps with shots and film alignment. This is so much faster!” Mike Pool, Field Service Specialist 3
International regulations recommend that the radiation and light field should be aligned within 2 % of the SID or even ± 5 mm at the chest wall for mammo. The RaySafe DXR+ sensor array has been specially designed to meet the needs of real-time checks of the light and radiation field alignment on radiographic and mammographic X-ray equipment.
The characteristics of the sensors provide an accurate and instantaneous result where adjustment is needed. A metric scale on the RaySafe DXR+ PCB is visible on the X-ray image.
1. Position the RaySafe DXR+
Position the "X-ray"-symbol within the light field.
Align the yellow center line with the light field edge
2. Expose and readout
Expose and read any X-ray/light field deviation
in the RaySafe DXR+ display.